
New TopShim run:

version = topshim 2.0.0
date = 2024-07-24
time = 03:27:04.055
command = topshim 

FINE SHIMMING

Parameters:

probe = Z108618_0624
solvent = CDCl3
shim nucleus = 2H
nucleus optimised for = 1H
o1p (from lock) = 7.24 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before=27.4 Hz, Target= 0.2 Hz, After= 0.5 Hz
Iteration  2: FWHM Before= 0.5 Hz, Target= 0.3 Hz, After= 1.2 Hz
Iteration  3: FWHM Before= 1.2 Hz, Target= 0.2 Hz, After= 0.3 Hz
Iteration  4: FWHM Before= 0.3 Hz, Target= 0.3 Hz, After= 0.6 Hz
Iteration  5: FWHM Before= 0.6 Hz, Target= 0.4 Hz, After= 0.5 Hz
Iteration  6: FWHM Before= 0.5 Hz, Target= 0.5 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.989

shim changes: 
Z	-649
Z2	-190
Z3	+151
Z4	-6689
Z5	-11885

duration = 3 min 0 sec
completed successfully
finished Wed Jul 24 03:30:02 2024


========================================================================
