
New TopShim run:

version = topshim 2.0.0
date = 2024-03-09
time = 08:47:43.823
command = topshim 

FINE SHIMMING

Parameters:

probe = Z108618_0624
solvent = CDCl3
shim nucleus = 2H
nucleus optimised for = 13C
o1p (from lock) = 7.24 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 1.0 Hz, Target= 0.5 Hz, After= 0.6 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 1.27

shim changes: 
Z	+21
Z2	-137
Z3	-1088
Z4	+551
Z5	+4394

duration = 1 min 15 sec
completed successfully
finished Sat Mar  9 08:48:57 2024


========================================================================
