
New TopShim run:

version = topshim 2.0.0
date = 2024-07-09
time = 12:15:17.290
command = topshim 

FINE SHIMMING

Parameters:

probe = Z168772_0022
solvent = CDCl3
shim nucleus = 2H
nucleus optimised for = 1H
o1p (from lock) = 7.24 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 0.5 Hz, Target= 0.4 Hz, After= 0.5 Hz
Iteration  2: FWHM Before= 0.5 Hz, Target= 0.3 Hz, After= 0.5 Hz
Iteration  3: FWHM Before= 0.5 Hz, Target= 0.5 Hz, After= 0.5 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.549

shim changes: 
Z	-7
Z2	-5
Z3	+4
Z4	+14
Z5	-7

duration = 2 min 18 sec
completed successfully
finished Tue Jul  9 12:17:34 2024


========================================================================
