
New TopShim run:

version = topshim 2.0.0
date = 2024-03-09
time = 09:59:58.340
command = topshim 

FINE SHIMMING

Parameters:

probe = Z108618_0624
solvent = CDCl3
shim nucleus = 2H
nucleus optimised for = 13C
o1p (from lock) = 7.24 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 1.9 Hz, Target= 0.4 Hz, After= 1.5 Hz
Iteration  2: FWHM Before= 1.5 Hz, Target= 0.9 Hz, After= 1.8 Hz
Iteration  3: FWHM Before= 1.8 Hz, Target= 0.5 Hz, After= 0.9 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 1.26

shim changes: 
Z	+42
Z2	-123
Z3	-1121
Z4	+477
Z5	+4691

duration = 2 min 13 sec
completed successfully
finished Sat Mar  9 10:02:10 2024


========================================================================
