
New TopShim run:

version = topshim 2.0.0
date = 2024-03-08
time = 16:38:10.420
command = topshim 

FINE SHIMMING

Parameters:

probe = Z108618_0624
solvent = CDCl3
shim nucleus = 2H
nucleus optimised for = 1H
o1p (from lock) = 7.24 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 0.6 Hz, Target= 1.5 Hz, After= 1.7 Hz
Iteration  2: FWHM Before= 1.7 Hz, Target= 0.9 Hz, After= 1.2 Hz
Iteration  3: FWHM Before= 1.2 Hz, Target= 1.1 Hz, After= 1.6 Hz
Iteration  4: FWHM Before= 1.6 Hz, Target= 1.6 Hz, After= 1.4 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 1.44

shim changes: 
Z	+70
Z2	-128
Z3	-987
Z4	+486
Z5	+3997

duration = 3 min 15 sec
completed successfully
finished Fri Mar  8 16:41:24 2024


========================================================================
