
New TopShim run:

version = topshim 2.0.0
date = 2024-03-11
time = 10:14:02.968
command = topshim 

FINE SHIMMING

Parameters:

probe = Z108618_0624
solvent = CDCl3
shim nucleus = 2H
nucleus optimised for = 19F
o1p (from lock) = 7.24 ppm

Results:

NOTE: No 19F B1-profile found. Optimizing shimming as if for NMR nucleus = 1H


1D Shimming:
Iteration  1: FWHM Before= 2.4 Hz, Target= 1.0 Hz, After= 1.8 Hz
Iteration  2: FWHM Before= 1.8 Hz, Target= 1.0 Hz, After= 1.5 Hz
Iteration  3: FWHM Before= 1.5 Hz, Target= 1.5 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 1.38

shim changes: 
Z	-13
Z2	-149
Z3	-764
Z4	+404
Z5	+2545

duration = 1 min 45 sec
completed successfully
finished Mon Mar 11 10:15:46 2024


========================================================================
