
New TopShim run:

version = topshim 2.0.0
date = 2024-04-12
time = 23:24:46.754
command = topshim 

FINE SHIMMING

Parameters:

probe = Z108618_0625
solvent = CDCl3
shim nucleus = 2H
nucleus optimised for = 13C
o1p (from lock) = 7.24 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 1.2 Hz, Target= 0.3 Hz, After= 0.5 Hz
Iteration  2: FWHM Before= 0.5 Hz, Target= 0.4 Hz, After= 0.4 Hz
Iteration  3: FWHM Before= 0.4 Hz, Target= 0.3 Hz, After= 0.3 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.454

shim changes: 
Z	-182
Z2	-118
Z3	-289
Z4	-658
Z5	+645

duration = 1 min 21 sec
completed successfully
finished Fri Apr 12 23:26:06 2024


========================================================================
