
New TopShim run:

version = topshim 2.0.0
date = 2024-05-01
time = 18:49:19.935
command = topshim 

FINE SHIMMING

Parameters:

probe = Z108618_0624
solvent = CDCl3
shim nucleus = 2H
nucleus optimised for = 1H
o1p (from lock) = 7.24 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 5.1 Hz, Target= 0.9 Hz, After= 1.5 Hz
Iteration  2: FWHM Before= 1.5 Hz, Target= 0.7 Hz, After= 1.8 Hz
Iteration  3: FWHM Before= 1.8 Hz, Target= 1.6 Hz, After= 0.5 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 1.06

shim changes: 
Z	-118
Z2	-87
Z3	-283
Z4	-67
Z5	+323

duration = 2 min 28 sec
completed successfully
finished Wed May  1 18:51:46 2024


========================================================================
