
New TopShim run:

version = topshim 2.0.0
date = 2024-04-12
time = 20:37:13.499
command = topshim 

FINE SHIMMING

Parameters:

probe = Z108618_0625
solvent = CDCl3
shim nucleus = 2H
nucleus optimised for = 13C
o1p (from lock) = 7.24 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 4.2 Hz, Target= 0.4 Hz, After= 0.2 Hz
Iteration  2: FWHM Before= 0.2 Hz, Target= 1.4 Hz, After= 2.7 Hz
Iteration  3: FWHM Before= 2.7 Hz, Target= 1.1 Hz, After= 1.5 Hz
Iteration  4: FWHM Before= 1.5 Hz, Target= 0.7 Hz, After= 0.8 Hz
Iteration  5: FWHM Before= 0.8 Hz, Target= 0.5 Hz, After= 0.6 Hz
Iteration  6: FWHM Before= 0.6 Hz, Target= 0.3 Hz, After= 0.6 Hz
Iteration  7: FWHM Before= 0.6 Hz, Target= 0.2 Hz, After= 0.5 Hz
Iteration  8: FWHM Before= 0.5 Hz, Target= 0.4 Hz, After= 0.5 Hz
Using the best result as the final shim state.

Lineshape quality factor (1D spectrum, spinning) = 1.33

shim changes: 
Z	+2691
Z2	-3315
Z3	+8238
Z4	-46996
Z5	+47571

duration = 2 min 54 sec
completed successfully
finished Fri Apr 12 20:40:06 2024


========================================================================
