
New TopShim run:

version = topshim 2.0.0
date = 2024-04-12
time = 18:55:06.060
command = topshim 

FINE SHIMMING

Parameters:

probe = Z108618_0625
solvent = CDCl3
shim nucleus = 2H
nucleus optimised for = 1H
o1p (from lock) = 7.24 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 0.3 Hz, Target= 0.2 Hz, After= 0.6 Hz
Iteration  2: FWHM Before= 0.6 Hz, Target= 0.3 Hz, After= 0.4 Hz
Iteration  3: FWHM Before= 0.4 Hz, Target= 0.3 Hz, After= 0.4 Hz
Iteration  4: FWHM Before= 0.4 Hz, Target= 0.3 Hz, After= 0.5 Hz
Iteration  5: FWHM Before= 0.5 Hz, Target= 0.3 Hz, After= 0.3 Hz
Iteration  6: FWHM Before= 0.3 Hz, Target= 0.3 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.398

shim changes: 
Z	-274
Z2	-197
Z3	-380
Z4	-1561
Z5	-586

duration = 1 min 51 sec
completed successfully
finished Fri Apr 12 18:56:56 2024


========================================================================
