
New TopShim run:

version = topshim 2.0.0
date = 2024-03-12
time = 17:45:38.566
command = topshim 

FINE SHIMMING

Parameters:

probe = Z168772_0022
solvent = CDCl3
shim nucleus = 2H
nucleus optimised for = 13C
o1p (from lock) = 7.24 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 0.3 Hz, Target= 0.2 Hz, After= 0.2 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.354

shim changes: 
Z	-58
Z2	-148
Z3	+40
Z4	-165
Z5	-48
Z6	-505
Z7	+626
Z8	+2916

duration = 38 sec
completed successfully
finished Tue Mar 12 17:46:15 2024


========================================================================

New TopShim run:

version = topshim 2.0.0
date = 2024-03-12
time = 17:46:27.774
command = topshim  tunea tune

TUNE

Parameters:

probe = Z168772_0022
solvent = CDCl3

Results:


Performing tune after shimming
tune lock level increase = 1.2

shim changes: 
Z	-6
X	+24
XZ	+325
Y	-1
YZ	+58

duration = 1 min 10 sec
completed successfully
finished Tue Mar 12 17:47:36 2024


========================================================================

New TopShim run:

version = topshim 2.0.0
date = 2024-03-12
time = 17:48:21.435
command = topshim 

FINE SHIMMING

Parameters:

probe = Z168772_0022
solvent = CDCl3
shim nucleus = 2H
nucleus optimised for = 13C
o1p (from lock) = 7.24 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 0.2 Hz, Target= 0.2 Hz, After= 0.8 Hz
Iteration  2: FWHM Before= 0.8 Hz, Target= 0.2 Hz, After= 0.4 Hz
Iteration  3: FWHM Before= 0.4 Hz, Target= 0.2 Hz, After= 0.2 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.333

shim changes: 
Z	+4
Z2	-7
Z3	-6
Z4	-64
Z5	+46
Z6	+344
Z7	-414
Z8	-3049

duration = 57 sec
completed successfully
finished Tue Mar 12 17:49:17 2024


========================================================================
