
New TopShim run:

version = topshim 2.0.0
date = 2024-05-01
time = 23:04:21.512
command = topshim 

FINE SHIMMING

Parameters:

probe = Z108618_0624
solvent = CDCl3
shim nucleus = 2H
nucleus optimised for = 13C
o1p (from lock) = 7.24 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 7.7 Hz, Target= 0.5 Hz, After= 1.5 Hz
Iteration  2: FWHM Before= 1.5 Hz, Target= 1.8 Hz, After= 1.7 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 1.7

shim changes: 
Z	-284
Z2	-237
Z3	-57
Z4	-714
Z5	-2345

duration = 1 min 45 sec
completed successfully
finished Wed May  1 23:06:05 2024


========================================================================
