
New TopShim run:

version = topshim 2.0.0
date = 2024-05-01
time = 18:53:50.923
command = topshim 

FINE SHIMMING

Parameters:

probe = Z108618_0624
solvent = CDCl3
shim nucleus = 2H
nucleus optimised for = 1H
o1p (from lock) = 7.24 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 7.2 Hz, Target= 1.3 Hz, After= 1.4 Hz
Iteration  2: FWHM Before= 1.4 Hz, Target= 1.3 Hz, After= 1.4 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 1.33

shim changes: 
Z	-238
Z2	-203
Z3	-334
Z4	-656
Z5	-625

duration = 2 min 5 sec
completed successfully
finished Wed May  1 18:55:54 2024


========================================================================
