
New TopShim run:

version = topshim 2.0.0
date = 2024-03-08
time = 16:43:24.699
command = topshim 

FINE SHIMMING

Parameters:

probe = Z108618_0624
solvent = CDCl3
shim nucleus = 2H
nucleus optimised for = 1H
o1p (from lock) = 7.24 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 3.6 Hz, Target= 0.3 Hz, After= 1.5 Hz
Iteration  2: FWHM Before= 1.5 Hz, Target= 1.5 Hz, After= 1.7 Hz
Iteration  3: FWHM Before= 1.7 Hz, Target= 1.7 Hz, After= 1.8 Hz
Iteration  4: FWHM Before= 1.8 Hz, Target= 1.7 Hz, After= 1.5 Hz
Iteration  5: FWHM Before= 1.5 Hz, Target= 1.5 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 1.64

shim changes: 
Z	-74
Z2	-206
Z3	-827
Z4	+24
Z5	+2017

duration = 2 min 49 sec
completed successfully
finished Fri Mar  8 16:46:12 2024


========================================================================
