
New TopShim run:

version = topshim 2.0.0
date = 2024-06-05
time = 19:32:51.404
command = topshim 

FINE SHIMMING

Parameters:

probe = Z108618_0624
solvent = CDCl3
shim nucleus = 2H
nucleus optimised for = 1H
o1p (from lock) = 7.24 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 0.2 Hz, Target= 0.4 Hz, After= 0.3 Hz
Iteration  2: FWHM Before= 0.3 Hz, Target= 0.3 Hz, After= 0.5 Hz
Iteration  3: FWHM Before= 0.5 Hz, Target= 0.5 Hz, After= 0.4 Hz
Iteration  4: FWHM Before= 0.4 Hz, Target= 0.3 Hz, After= 0.5 Hz
Iteration  5: FWHM Before= 0.5 Hz, Target= 0.4 Hz, After= 0.9 Hz
Iteration  6: FWHM Before= 0.9 Hz, Target= 0.3 Hz, After= 0.7 Hz
Iteration  7: FWHM Before= 0.7 Hz, Target= 0.4 Hz, After= 0.7 Hz
Iteration  8: FWHM Before= 0.7 Hz, Target= 0.3 Hz, After= 0.4 Hz
Iteration  9: FWHM Before= 0.4 Hz, Target= 0.3 Hz, After= 0.9 Hz
Using the best result as the final shim state.

Lineshape quality factor (1D spectrum, spinning) = 0.542

shim changes: 
Z	-568
Z2	-365
Z3	-274
Z4	-3123
Z5	-5151

duration = 5 min 9 sec
completed successfully
finished Wed Jun  5 19:37:59 2024


========================================================================
