
New TopShim run:

version = topshim 2.0.0
date = 2024-05-01
time = 22:26:29.629
command = topshim 

FINE SHIMMING

Parameters:

probe = Z108618_0624
solvent = CDCl3
shim nucleus = 2H
nucleus optimised for = 13C
o1p (from lock) = 7.24 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 6.1 Hz, Target= 1.2 Hz, After= 1.3 Hz
Iteration  2: FWHM Before= 1.3 Hz, Target= 0.9 Hz, After= 1.1 Hz
Iteration  3: FWHM Before= 1.1 Hz, Target= 1.0 Hz, After= 1.4 Hz
Iteration  4: FWHM Before= 1.4 Hz, Target= 0.4 Hz, After= 1.6 Hz
Iteration  5: FWHM Before= 1.6 Hz, Target= 1.4 Hz, After= 1.3 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 1.48

shim changes: 
Z	-152
Z2	-88
Z3	-226
Z4	-51
Z5	-356

duration = 3 min 11 sec
completed successfully
finished Wed May  1 22:29:39 2024


========================================================================
