
New TopShim run:

version = topshim 2.1.0
date = 2024-09-11
time = 18:33:34.775
command = topshim 

FINE SHIMMING

Parameters:

probe = Z154039_0001
solvent = CDCl3
shim nucleus = 2H
nucleus optimised for = 13C
o1p (from lock) = 7.24 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before= 0.7 Hz, Target= 0.2 Hz, After= 0.2 Hz
Iteration 2: FWHM Before= 0.2 Hz, Target= 0.2 Hz, After= 0.3 Hz
Iteration 3: FWHM Before= 0.3 Hz, Target= 0.3 Hz, After= 0.2 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.315

shim changes: 
Z	-254
Z2	-44
Z3	+26
Z4	-111
Z5	-130
Z6	-132
Z7	+1416
Z8	+306

duration = 1 min 0 sec
completed successfully
finished Wed Sep 11 18:34:33 2024


========================================================================
